Hydrostatic pressure is the pressure that is generated by the weight of liquid above a measurement point, when the liquid is at rest. The height of a liquid column, of uniform density, is directly proportional to the hydrostatic pressure.
The hydrostatic properties of a liquid are not constant and the main factors influencing it are the density of the liquid and the local gravity. Both of these quantities need to be known in order to determine the hydrostatic pressure of a particular liquid.
View detailed information pages for instrumentation used to measure absolute pressure.
- Hydrostatic Level Sensors - Hydrostatic pressure sensing devices for measuring liquid level and converting it to an electrical signal to send to other instrumentation.
- Chemical Resistant Hydrostatic Liquid Level & Pressure Sensors - Pressure transmitters & transducers that are resistant to chemical corrosion for measuring the hydrostatic level of storage tanks or process pressures.
- 4-20mA Output Hydrostatic Level Transmitters & Sensors - Liquid level transmitters with a 4 to 20 milliamp current loop output signal for determining liquid depth or the fluid level by measuring the hydrostatic pressure.
- Underwater Pressure Sensors - Pressure sensors for installing underwater which are constructed to prevent water entering sensitive components and causing damage to the sensor.
- Submersible Hydrostatic Level Sensors & Probes - Level sensors for submerging permanently in liquid to determine the level or depth of fluid by measuring the hydrostatic pressure.
- Combined Temperature and Hydrostatic Liquid Level Sensors - Combined temperature & level sensors which provide a signal output for both temperature and hydrostatic pressure measurement from one device.
- SDI-12 Hydrostatic Liquid Level Sensors - SDI-12 interface hydrostatic level sensors with a serial digital interface for measuring liquid depth and returning the reading to low powered telemetry instrumentation in remote locations.
- Small Diameter Borehole Hydrostatic Liquid Level Sensors - Hydrostatic level sensors for measuring ground water level in small diameter bore-holes.
- Seawater Hydrostatic Liquid Level Sensors - Hydrostatic level sensors which are suitable for immersing in seawater for long periods to monitor depth and tidal changes.
- Absolute Unvented Reference Hydrostatic Liquid Level Sensors - Non-vented absolute range hydrostatic level sensors for measuring liquid level where it is not possible or practical to connect a vented gauge reference.
- Marine Approved Hydrostatic Liquid Level Sensors - Marine approved hydrostatic liquid level sensors for use in marine environments.
- Freshwater Hydrostatic Liquid Level Sensors - Liquid level sensors for measuring the level of freshwater in tanks, boreholes, reservoirs, rivers, canals, streams and wells.
- Plastic Submersible Hydrostatic Liquid Level Sensors - Chemical resistant submersible level sensors with a PVC housing and ceramic diaphragm for measuring the hydrostatic level of harsh chemicals.
- Plastic Externally Fitted Hydrostatic Liquid Level Sensors - Plastic liquid level sensors with a PVDF housing and Ceramic diaphragm for fitting to the outside of a chemical storage tank.
- Brackish Water Hydrostatic Liquid Level Sensors - Brackish water depth and level sensors with housing material for use with tidal and coastal sea water.
- Hazardous Area Externally Fitted Hydrostatic Liquid Level Sensors - Hazardous area external mount level sensors for use in zones where there is a high risk of explosive gases being present.
- Hazardous Area Submersible Hydrostatic Liquid Level Sensors - ATEX/IECEx intrinsically safe IP68 sensors for liquid level measurement in intrinsically safe installations on petrochemical, pharmaceutical, landfill and sewage treatment sites.
- 0-10 Volts Output Hydrostatic Sensors, Probes & Transducers - Analog output hydrostatic sensors with a 0 to 10 volt amplified output signal for measuring liquid head pressure or fluid level.
- Water Resistant Pressure Sensors with Strain Gauge Output - Water resistant strain gauge output pressure sensor for measuring hydrostatic liquid level or underwater system pressures.
- Hydrostatic Liquid Level Voltage Output Pressure Transducers - Hydrostatic pressure transducers for measuring the height of a liquid column via a proportional amplified voltage output signal.
- Hydrostatic Pressure Transmitters - Current loop output sensor transmitters for measuring the hydrostatic pressure exerted by a static head of liquid.
- Hydrostatic Liquid Level Switches - Hydrostatic pressure sensing switches for activating switch contacts at predetermined liquid level thresholds.
Formula for calculating hydrostatic pressure
The formula for calculating the hydrostatic pressure of a column of liquid is:
Phyd = h·ρ·g
Prel = h·ρ·g
Pabs = h·ρ·g + Patm
- Phyd = Hydrostatic Pressure (Pa)
- Prel = Relative Pressure (Pa)
- Pabs = Absolute Pressure (Pa)
- Patm = Atmospheric Pressure (Pa)
- h = Liquid Height (m)
- ρ = Liquid Density (kg/m3)
- g = Local Gravity (m/s2)
How atmospheric pressure affects hydrostatic pressure measurement
Since hydrostatic pressure is the pressure solely generated by a column of liquid, it maybe the case that the pressure measured includes the additional component of atmospheric pressure acting on the liquid surface.
In the cases of measuring absolute pressure it is necessary to also measure and subtract the atmospheric pressure to determine the true hydrostatic pressure.
How local gravity affects hydrostatic pressure
The local gravity depends on latitudinal position and height above sea level.
For convenience the most popular pressure units for measuring hydrostatic pressure are metres of water and feet of water at 4 deg C (39.2 degF) with a standard gravity of 9.80665 m/s2. The density of pure water at 4 deg C is very close to 1000 kg/m3 and therefore this has been adopted as the standard density of water. Another reason for the significance of choosing 4 deg C is that it is very close to the temperature that water reaches its maximum density.
How temperature affects hydrostatic pressure
The density of a liquid will vary with changes in temperature so this is often quoted alongside hydrostatic pressure units e.g. mH2O @ 4 deg C.
In practical terms hydrostatic pressure units are rarely absolutely precise because the temperature of any liquid is not always going to be 4 deg C.
You will also come across another temperature standard of 60 deg F (15.56 deg C). This can lead to confusion and inaccuracies when the temperature is not labelled alongside the hydrostatic pressure unit.
For most applications the different temperature dependent unit conversions are not significant enough to influence the results, since the reading accuracy is often much wider than the difference in the pressure unit conversion factor at these 2 temperatures.
In summary hydrostatic pressure units are a very convenient method for relating pressure to a height of fluid but they are not absolute pressure units and it is not always clear what density/temperature has been assumed in their derivation, so be very cautious when using them for high precision level measurements. In fact some institutions are discouraging their use because of the very reasons mentioned above and recommending the use of fixed value pressure units instead of units that are open to interpretation and are dependent on gravity or temperature.
Questions & Answers about hydrostatic pressure
Bubble entrainment affects on level and pressure reading
We are measuring water level in an open tank using a vented pressure gauge. When the liquid is discharged into the tank, bubbles are entrained. The bubbles then float to the surface and dissipate. The additional volume of bubbles increases the level of the water in the tank but not the volume of water in the tank. Can we expect a pressure sensor mounted in the bottom of the tank to read the same value whether there are bubbles or not? Or put another way will the vented sensor read the pressure generated by the water alone whether the compensated atmosphere is entirely above the water or partially entrained in the water?
If you consider the formula for calculating hydrostatic pressure, the main variables in your application are density and height. Since the density will drop by the same proportion as the increase in water surface height, they will both cancel each other out. Therefore the measured pressure will remain the same irrespective of the change in visual liquid level due to bubble entrainment.
Pressure generated by gravity fed water system in a house
How many bars pressure does a gravity fed water system generate? Our house has the water header tank located in the loft. The bathroom is one floor below the cold and hot water tanks.
The pressure generated by a vented water tank is equal to the difference in water height between the water surface inside the tank (typically near the top when there is no demand) and the measurement point. Since the pressure generated is dependent upon the height of water above the measurement point, the pressure will change at different levels within the house.
- 1 metre of water is approximately equal to 0.098 bar, so the difference in height (m) between the vented/open header water tank and the water tap (faucet) multiplied by 0.098 will equal the pressure in bar.
- 1 foot of water is approximately equal to 0.03 bar, so the difference in height (ft) between the vented/open header water tank and the water tap (faucet) multiplied by 0.03 will equal the pressure in bar.
Temperature of liquid
Why is temperature included with height of fluid pressure units?
Pressure units which are related to the height of a liquid often include a reference temperature e.g. 50 mH2O @ 4degC.
The hydrostatic pressure of a fluid at a certain height is determined by the total height of fluid above that point. If the temperature of the liquid is increased it will expand in volume, thus increasing the fluid level. However the hydrostatic pressure will remain the same, therefore when quoting a pressure in terms of fluid level it is useful to know the temperature.
For example a tank full of water located above ground level on a warm day in the UK might be 28 degC during the middle of the day, whereas during the early hours of the morning it might drop as low as 8 degC. The difference in level due to a 20degC change would be approximately 0.2% without any change in pressure. This does not seem much but when you consider that many pressure sensors can measure to a better precision than 0.25% of full scale a 0.2% change in water density is quite significant.
For applications where you are interested in the weight of the tank contents rather than the volume, a fluid level reading which is independent of changes in density due to temperature variation can actually be very useful.
A hydrostatic pressure unit can be specified for any reference temperature, but to simplify calibration and establish commonality across manufacturers, standardised temperatures are used such as 4 degC (39.2 degF) and 60 degF (15.56 degC).
Many manufacturers do not explain on their product data sheets which temperature they have used, particular with sensors where absolute accuracy is not so important, so if high accuracy is important to your application, calibration should be verified during installation.
Actual liquid height vs liquid height pressure units
How well do liquid height derived pressure units such as mH2O & ftH2O relate to the actual liquid height measured in metres & feet?
Since liquid height derived pressure units have a fixed value, they rarely match the real height measurement of a liquid. This is because the hydrostatic pressure generated by a measured height of liquid will vary depending on the liquid temperature and local gravity.
- Using absolute pressure sensors to measure hydrostatic level
- How to reduce submersible pressure sensor failures caused by moisture ingress
- Why are Submersible applications so demanding of Pressure Transducers?
- Considerations for monitoring Landfill Water Level
- How does changes in Local Gravity affect Hydrostatic Pressure
- Improve the electrical termination of vented submersible pressure sensors
- Measuring liquid level in a sealed tank with a hydrostatic pressure sensor
- Determining the hydrostatic pressure range for a tank level sensor
- Using a screw-in or waterproof pressure transmitter to measure liquid level
- Difference between vented and non-vented water level measurement
- Determining water tank volume using hydrostatic pressure
- Measuring the volume of liquid in a horizontal cylinder tank
- Hydrostatic Liquid Level Sensors
- How do you measure the volume of a liquid using pressure
- Calculating required pressure range for fuel tank level measurement
- cmH2O – Centimetres of Water Column at 4 deg C Pressure Unit
- ftH2O – Feet of Water Column at 4 deg C Pressure Unit
- mH2O – Metres of Water Gauge at 4 deg C Pressure Unit
- mmH2O – Millimetres of Water Column at 4 deg C Pressure Unit
- Specific Gravity