Click to reload page with unique web address for bookmarking or sharing current settings

**Hydrostatic liquid level measurement products**

- Hydrostatic Level Sensors
- 4-20mA Output Hydrostatic Level Transmitters & Sensors
- LMK387 Cleanable Sewage and Sludge Pressure/Level Sensor
- Underwater Pressure Sensors

## Related Tools

- Fluid depth to hydrostatic pressure calculator
- Fluid depth, density, gravity and pressure calculator
- Local gravity calculator
- Pressure units converter
- Length units converter

## User Guide

This calculator and conversion scale is used to determine the height of a column of liquid from the pressure generated at the bottom of the liquid column and show a custom pressure to liquid depth conversion scale.

#### Formulas

The calculation formulas used for this tool are:

L = P / (ρ · g)

ρ = ρ_{0} · SG

L = P / (ρ_{0} · SG · g)

###### Symbols

- L = Liquid height
- P = Pressure
- g = local gravity (e.g. standard = 9.80665ms
^{-2}) - ρ = Density of liquid
- ρ
_{0}= Density of fresh water (1000 kgm^{-3}@4°C) - SG = Specific gravity of liquid (e.g. fresh water = 1)

#### Hydrostatic Pressure

Enter the pressure reading measured at the base of the fluid column. It is assumed that the pressure value is the difference in pressure between the measurement point and the top of the fluid. In most applications this will be the measured pressure since the top of the fluid column is vented to atmosphere, and the pressure is typically measured in relation to atmospheric pressure. If the top of the liquid column is sealed, it will be necessary to detemine the pressure difference to correct for any positive or negative pressure acting on the liquid surface.

#### Specific Gravity (SG)

Enter here the ratio of liquid density compared to the density of fresh water (1000 kg/m^{3}). The default value is set to an SG of 1.00, which corresponds to fresh water at 4 degrees Celsius.

#### Local Gravity (g)

Enter the acceleration due to gravity for your geographical location in metres per second per second (ms^{-2}). The local gravity is dependent on several factors such as latitude, height above sea-level, local geological density, etc… Refer to your national geological survey data for your location or use this local gravity calculator to determine a close approximation. The default value is set to 9.80665 ms^{-2} which is standard gravity.

#### Height of Liquid Column

This is the calculated height of fluid above the point at which the pressure is measured.

## Help

### Pressure to diesel fuel level conversion

*What parameters are required to convert a pressure reading into diesel fuel level?*

The level of diesel fuel in a tank is one of the dimensions used to calculate the volume of diesel fuel. If you know the density and weight of the diesel, then you can calculate the height or level of diesel fuel. For a tank with uniform vertical shape such as a vertical cylinder or a rectangular one, this calculation is very straightforward. The pressure reading is directly proportional to the weight of diesel above the sensor, so the level is calculated by first multiplying the density and acceleration due to gravity together, and then dividing the result into the pressure reading.

When specifying a pressure sensor to measure diesel level, calculate the required pressure range using the maximum required diesel level and specific gravity of diesel. The pressure to liquid level calculator on this page will do the calculation for you and there is more explanation of the maths involved in the above user guide.

**Hydrostatic liquid level measurement products**

- Underwater Pressure Sensors
- LMK387 Cleanable Sewage and Sludge Pressure/Level Sensor
- LMK382 Low Range IP68 Waste Water Level Transmitter
- DPT200 Pressurised Tank Level Differential Pressure Transmitter