What is the method of shunt calibration in relation to an indicator connected to a pressure transducer?
Shunt calibration is a convenient way of calibrating a wheatstone bridge strain gauge circuit which is a common technology used in pressure transducers. It’s not a complete calibration, since no actual pressure is applied when conducting a shunt calibration, but it does reduce the errors associated with the electrical characteristics of the strain gauges and the connected electrical components.
Basically a shunt resistor is added in parallel to one of the strain gauges to simulate what would happen if a real load was measured by the pressure transducer. Since the resistance is known, the resulting output in millivolts can be compared to what should be expected when the equivalent real pressure is applied. Then the difference in output signal can be compensated within the monitoring instrumentation.
- DMP334 Hydraulic Pressure Transducer for Very High Pressures
- 1 bar absolute strain gauge mV out pressure sensor for educational instrumentation
- 40B48 Strain Gauge Indicator with Relay Switch Alarms
- 40B96 Strain Gauge Readout with Process Alarm Relays
Request info on strain gauge related products for your application.