I have a confusion regarding the difference between the maximum working pressure, overpressure and burst pressure for pressure transmitters.
Max working pressure is the operating range of the pressure sensor, normally described as the “full scale” or “pressure range”.
Overpressure is the maximum amount of pressure or multiples of the measurement range that can be applied to a sensing device without causing damage. In some cases an allowance for performance degradation will be included in the overpressure rating, but since this is a typically a zero offset, it can be easily removed by adjustment to the electronics.
Overpressure is sometimes mistaken for burst pressure which is the maximum safe pressure or multiple of the measurement range that can be applied without causing a breach of pressure containment.
e.g. A pressure transmitter with a 7 bar g pressure range described as having a 4x overpressure and a 10x burst pressure would have the following pressure limits:
- Full Scale = 7 barg
- Overpressure Rating = 28 barg
- Burst Pressure = 70 barg
- Measure filter dp to 20mbar & withstand 1bar cleaning pulse
- Single differential versus twin absolute pressure sensor for measuring dp
- 100 mbar abs vacuum transmitter with a 5 bar overpressure
- DP sensor for -1 to 3 bar differential water pressure and 20 bar overpressure
Request info on overpressure protected products for your application.