Span Offset is the amount of deviation in output or reading from the exact value at a point within the measurement span.
Manufacturers may include a factory setting tolerance for the span offset on the product technical data sheet to help users determine the total error to be expected without any calibration adjustment.
For example a pressure transducer for measuring 0 to 15 psi pressure with a 0 to 30 mV output may have a span offset tolerance of +/-0.1% span. Therefore at 7.5 psi the maximum span setting tolerance will be +/- 0.015 mV.
- SCU for Isolating signal from sensors and trimming zero and full span
- 41X High Precision Low Range Pressure Transmitter
- 33X High Accuracy Digital Output Pressure Sensor
- Zero & Span calibration adjustable signal output pressure transducers
Request info on span output adjustable measurement products for your application.
Calibration Related Terms
More pages related to calibration technical terms.